Electromagnetic flowmeter with non-insulation pipe wall Online publication date: Wed, 01-Mar-2017
by Xuejing Li
International Journal of Computer Applications in Technology (IJCAT), Vol. 55, No. 2, 2017
Abstract: The internal pipe wall of the Electromagnetic Flowmeter (EMFM) must be non-conductive to prevent the generated electromotive force from short-circuiting. Usually the inside of metallic pipes is lined with insulating material. The lining limits the applicable temperature range of measured fluid and also its reliability. A new structure is proposed, in which the insulating liner is eliminated and replaced with non-insulation material. A potential distribution is formed on the pipe wall by applying voltage proportional to fluid flow rate. The potential distribution is decided by the conducting pipe wall. In this paper theoretical analysis and experimental results of EMFM with non-insulation pipe wall are described.
Online publication date: Wed, 01-Mar-2017
If you are not a subscriber and you just want to read the full contents of this article, buy online access here.Complimentary Subscribers, Editors or Members of the Editorial Board of the International Journal of Computer Applications in Technology (IJCAT):
Login with your Inderscience username and password:
Want to subscribe?
A subscription gives you complete access to all articles in the current issue, as well as to all articles in the previous three years (where applicable). See our Orders page to subscribe.
If you still need assistance, please email email@example.com